THE TISSUE COMPRESSION OPTIMIZATION AS THE CONDITION OF CONTROLLED IMPEDANCE CHANGE DURING ELECTRIC WELDING INTESTINAL ANASTOMOSIS CREATING
Introduction. The intestinal wall' structural unevenness determines uneven structure of the electric welding compound, and, accordingly, needs for increased compression to homogenize the density and electrical conductivity for better connection result. But at the same time, the required compression amount is close to destructive for intestine wall, since smaller quantities are ineffective.
Aim: was to create the optimum method for balancing and distributing the compression effort on intestinal tissue in the model of electro-welded anastomosis in order to achieve by this way the high frequency welding process stability during the gut walls connecting.
Material and methods. We performed 348 studies of programmed influence to pig organ complex the radio frequency electric voltage up to 200 V, with controlled change of impedance, after a pressure of 2.0 N/mm2 to 3.9 N/mm2 have applied on electrodes. The intestines were doubled in the instrument prototype, simulating an anastomosis. Electrode types were changed: smooth 247 mm2 (group 1); relief 280 mm2 (group 2); smooth 254 mm2 included limiters of their approach (group 3).
Results. In group 1 we tried to find by the steps of 0.1 N/mm2, the pressure value as a point of equilibrium of mechanical and electrical properties. In 7.5% studies the failure of basic impedance parameters achieving was observed, in 42.5% — significant deviation of growth rate, in 7.5% — its critical decrease. In group 2, the prevention of possible electrodes displacement was investigated by applying that's different relief configurations. In this series, the basic impedance reduction parameters achieving increased to 82.5%, but in the next phase its sudden drop was observed in 30% probes. Also in 32.5% the significant impedance fluctuations were observed. In group 3, the electrodes approaching limitations in the range of 0.03 mm to 0.3 mm were studied. The limiters usage proved to be most effective at the ratio: limiter height to tissues thickness between the electrodes in anastomosis model — as 1:40. A critical impedance drop was observed in only 2.5% of studies.
Conclusions. The electrodes pressure reducing increases the tissues impedance deviation from the target. The relief electrodes increases electrical conductivity and improves the starting part of electric welding, but increases the unevenness in its end part due to the impedance fluctuation. The limiters of electrodes approach usage provide the effect of full-time electric welding algorithm elaboration, regardless the values of optimal or subcritical intestinal compression.
Discussion. The obtained data are intended to serve as the technological solutions basis in creating the elements of the specialized electric weld anastomosis making tool. Their technological introduction will provide the further increase of yet established benefits of electric-welded anastomosis over sewn and stapled. This technological device could be also effective in other biological tissues welding.
Podpriatov, S. S., Podpryatov, S. E., Makarov, A. V., Marinsky, G. S., Tkachenko, V. A., Chernets, O. V., ... & Lopatkina, K. G. (2018). Vstanovlennya pervinnikh vimog do eksperimental'nikh zasobiv doslidzhennya ta umov stvorennya elektrozvarnogo z'ednannya stinok kishechniku [Establishing the first requirements in experimental equipment for investigations and creation conditions of electric welding intestinal connection]. Shpital'na khirurgiya (Hospital Surgery). Journal named by L. Ya. Kovalchuk, 2, 56-60. doi 10.11603/24144533.2018.2.9230. [In Ukrainian].
Podpriatov, S.S., Podpryatov, S. E., Marinsky, G. S., Chernets, O. V., Tkachenko, V. A., Bielousov, I. O., ... Serdyuk, V. K. (2017). Modelne eksperymentalne doslidzhennya mekhanichnoyi rezystentnosti stinky kyshky lyudyny pid tyskom v prototypi elektrozvaryuvalnoho instrumentu [Model experimental research of human intestine walls mechanical resistance under the pressure inside biologic welding instrument prototype]. Problemy viyskovoyi okhorony zdorovya (Problems of military health). 2, 325-331. [In Ukrainian].
Paton, B. E., Tkachenko, V. A., Marins'kii, G. S., Podpryatov, S. E., Chernets', O. V., Chvertko, N. A., ... & Matviichuk, G. M. (2014). Sposib z'ednannya zvaryuvannyam biologichnikh tkanin lyudei i tvarin z vikoristannyam visokochastotnogo strumu [The method of connecting the welding of biological tissues of people and animals using high-frequency current]. Ukraine Patent No. 106513. Bul. No. 17. [In Ukrainian].
J0rgensen, C. S., Dall, F. H., Jensen, S. L. & Gregersen, H. (2013). A new combined high-frequency ultrasound-impedance planimetry measuring system for the quantification of organ wall biomechanics in vivo. Int. Wound J., 10(4), 411-417. doi: 10.1016/0021-9290(95)95275-A.
Holmer, C., Winter, H., Kröger, M., Nagel, A., Jaenice, A., Lauster, R., ... & Ritz, J.P. (2011). Bipolar radiofrequency-induced thermofusion of intestinal anastomoses — feasibility of a new anastomosis technique in porcine and rat colon. Langenbecks Arch. Surg., 396(4), 529-533. doi: 10.1007/s00423-011-0756-0.
Sanchez-De Pedro, F., Moreno-Sanz, C., Morandeira-Rivas, A., Temas-Burillo, J. M., & Alhambra-Rodriguez De Guzman, Cr. (2014). Colorectal anastomosis facilitated by the use of the LigaSure® sealing device: comparative study in an animal model. Surg. Endosc., 28(2), 508-514. doi: 10.1007/s00464-013-3194-y.
Bosmans, J. W., Jongen, A. C., Bouvy, N. D., & Derikx, J. P. (2015). Colorectal anastomotic healing: why the biological processes that lead to anastomotic leakage should be revealed prior to conducting intervention studies. BMC Gastroenterology, 15(1), 180. doi: 10.1186/s12876-015-0410-3.
Han, S., Cai, Z., Ning, X., He, L., Chen, J., Huang, Z., ... Li, Z. (2015). Comparison of a new high-frequency electric welding system for intestinal closure with hand-sewn in vivo pig model. J. Laparoendosc. Adv. Surg. Tech. A, 25(8), 662-667. doi: 10.1089/lap.2015.0101.
Ho, Y.-H., & Ashour, M. A. T. (2010). Techniques for colorectal anastomosis. World Journal of Gastroenterology, 16(13), 1610-1621. doi.org/10.3748/ wjg.v16.i13.1610.
Journal Medical Informatics and Engineering allows the author(s) to hold the copyright without registration
Users can use, reuse and build upon the material published in the journal but only for non-commercial purposes
The majority of Medical Informatics and Engineering Open Access journals publish open access articles under the terms of the Creative Commons Attribution (CC BY) License which permits use, distribution and reproduction in any medium, provided the original work is properly cited. The remaining journals offer a choice of licenses.